Where do I send the manual for Alcon Constellation vision system?
Portable and mobile RF communications equipment can affect this medical electrical equipment. 8065751025 ix fConstellation Vision System Comments or corrections concerning this manual should be addressed to: Alcon Technical Services Group PO BOX 19587 Irvine, CA, USA 92623 All rights reserved.
What is the output voltage of Alcon fconstellation vision system?
OUTPUT CONTROL SETTING Note: Maximum output peak-to-peak voltage is about 140V without resistive load. 1.20 8065751025 fConstellation Vision System PRODUCT SERVICE For product service, please contact Alcon’s Technical Services Department at the number provided below.
What are the guidelines for use of constellation® vision system?
? Good clinical practice dictates the testing for adequate irrigation, aspiration ?ow, and operation as applicable for each handpiece prior to entering the eye. ? Do not use the Constellation Vision System system near ?ammable anesthetics. ? Use only Alcon-supplied A.C. power cords.
How do I contact Alcon for installation?
installation at (800) 832-7827. Outside the USA contact your local Alcon af?liate. source air pressure. The system operates automatically with pressures of 58 (4 bar) to
What does advisory message mean?
Advisory Messages indicate a minor failure in the system, typically a situation that can be corrected by the user. When an advisory is detected, the following actions occur:
What is Table 5-1?
Table 5-1 is a general troubleshooting guide that addresses symptoms/observations and what the operator can do to try and solve the observed problem. Figure 5-5 and Table 5-2 are presented as aids to rapid location of failed or malfunctioning parts or components in the Constellation® Vision System; they are not meant to replace standard troubleshooting methods. In all cases, should the corrective actions not provide the desired result, call your Alcon Technical Services Department.
What are Constellation consumables?
Constellation® Consumables Procedure Paks are available in multiple configurations to meet the user’s needs for each procedure. There are three main types of Consumables Paks: Posterior segment, Anterior segment and Combined Procedure (both Posterior and Anterior), and each type of pak can be customized further to specify items such as gauge specific instruments.
What is a cassette in a Constellation console?
The cassette is the interface between the Constellation® console and the surgical handpiece. It is used to regulate BSS® irrigating fluid to the handpiece, aspirate debris from the handpiece, monitor irrigation and aspiration pressure, and deposit the debris in a sealed drainage bag for disposal.
Where are system error messages displayed?
System Error Messages are displayed in a dialog box as shown in Figure 5-2 System Errors require a partial system shutdown. When an error is detected, the following actions occur.
Does Alcon use enzymatic cleaners?
Due to the potential for Toxic Anterior Segment Syndrome (TASS), Alcon does not recommend the use of enzymatic cleaners and detergents. If however, local jurisdictions mandate their use relative to ophthalmic instruments, the materials of construction are compatible, up to a pH of 11.3, when the enzymatic chemicals and detergents are completely rinsed/neutralized immediately after cleaning/processing.